
PCEPA 6000 power meter 

The PCEPA 6000 power meter is a tabletop device for measuring effective power, apparent power, power factor, energy consumption, current, continuous voltage, resistance and frequency. The power meter also offers the possibility to connect industrial current converters with a transmission relation of 100/5A or 1000/5A. Its large 93 x 52mm LCD is easy to read and provides a variety of results, including Watts, cos φ, ACV and ACA. With the optional PCEPA ADP current adaptor with ground wire (up to 16ACA) it can safely and quickly measure and analyse with the help of the power meter. Measurement of current and power can be done directly with test cables that come with the device or indirectly using the clamps. The device has an RS232 interface, and software to analyse results and transfer data to a computer. This makes it the ideal device for any workshop or for measuring power to determine faults in devices. This link shows an overview listing each power meter we have available to suit your needs 

 Multimeter function  Measurement of direct current connecting a sensor head or current clamp  Measures effective and apparent power  Measures power factor  Measures energy consumption (kWh)  Data hold / Peak hold  Function Hold / Alarm function High / Low  Interface RS232  Powered by mains adaptor or batteries  Safety standards: IEC10101; CAT II 600V 



Amp clamp adaptor for measuring current indirectly up to 1000A without interrupting the conductor. 
Amp clamp adaptor for the PCEPA 6000 power meter 


Technical specifications  
Function  Measurement range; Accuracy; Resolution 
Measurement of real power (AC, real value of power), direct measurement  1 W ... 6 kW; 1 W; ±1,5 % + 1 dgt. 
Measurement of real power (AC, real value of power), with adaptor ext. or current converter 
1 W ... 999,9 kW; 1 W; ±1,5 % + 1 dgt. 
Measurement of apparent power (AC) 
0.01VA to 9.999kVA; 0.01VA; ±1.5% +1 digit 
Power factor (cos φ)  0.00 to 1.00; 0.01; ±1.5% +2 digit 
KWh counter 
0.001Wh to 9,999kWh; 0.001Wh; ±1% +1 digit 
ACV / DCV  0.1V to 600V; 0.1 / 1V; ±1% +1 digit 
ACA eff/rms 
direct maximum 10A; 10mA, with induction head max. 1,000A;1A; ±1% +1 digit 
ACA with clamps  up to 2,000A (depending on clamps) 
Frequency  10.0 to 999Hz; 0.1Hz; 1% +1 digit 
Ohms  1 Ω to 19.99kΩ; 1 / 10 Ω; ±1% +1 digit 
Display  93 x 52mm multi functional LCD 
Operating conditions  max. 80% r.h. / 0 to 50°C 
Power  6 AA batteries or mains adaptor 
Dimensions  280 x 210 x 90mm 
Weight  1,600g 
Examples of use  
In the following image there is a PCEPA 6000 being used to repair a computer. Using the current adaptor, it can quickly take measurements of consumption and measurements of power without the need to use too many cables. 

Here is an image of the power meter's large LCD that can show four different readings simultaneously. The display mode depends on the function setting. 

In this instance, the device is being used to test the thermal value of an electronic thermal ventilator once repaired with our PCEPA 6000 power meter. The usefulness of the current adaptor can be appreciated here, and it is easy to attach safely without the mess of external cables. With the power meter, tests can be done easily, safely and accurately an all types of small electrical appliances up to 16A, including high consumer electronics. 

Contents 

Optional accessories  Adaptor for current to measure grounded plugs quickly and safely.  Amp clamps for measuring inductively driven current up to 1,000A  Laboratory calibration containing documentation and ISO certificate  Main adaptor for stationary use 



Here you can find other similar products under the classification of "power meter":  PCEPA 8000 series power meter  PCEUT232 series power meter  PCEGPA 62 series power meter  PCE360 series power meter  PCE830 series power meter 

Here you will find all the measuring instruments available at PCE Instruments.  
Contact: PCE Instruments UK Limited Units 12/13 Southpoint Business Park Ensign Way, Southampton United Kingdom, SO31 4RF Phone. +44 ( 0 ) 2380 98703 0 Fax +44 ( 0 ) 2380 98703 0 

This page in German in Italian in Spanish in Croatian in French in Dutch in Hungarian in Polish in Serbian in Russian in Turkish in Portuguese in Portuguese 
