electronic crane scale is a precise device. Within electromagnetic fields can
lead to large aberrations. In this case the location of the crane scale must be
changed. Disturbing surrounding conditions, e.g. air flow or vibrations should
be avoided. Also avoid quick temperature changes, or perform new calibration
after adjustment to environment. Avoid high humidity, damp and dust, since the
crane scale is not hermetically sealed. Protect the crane scale from liquids,
since they might penetrate the measuring system. Thus do only clean the crane
scale with dry or slightly damped equipment. Do not use any solution since
they might damage the lacquer or any plastic parts. Remove spilled measuring
goods immediately. A short heat-up time after starting the device leads to more
precise results. Please consider a careful operation with the crane scale. Do
not hang any continuous load exceeding the operation range of the scale.
Strictly avoid any impact or overload exceeding the maximum load (max), since it
might damage the scale. If there are any errors in the program, restart the
scale. In that case the operation program must be restarted. Do not install the
scale in explosive interiors, since the standard model is not Ex-proved. Test
the crane scale with calibration weights on a regular basis.
guarantee claim is revoked as soon as the crane scale is demounted in any way or
the application of the crane scale exceeds the listed specifications.
keep package for any potential return. Do only use the original package for
PCE Instruments UK Limited
Unit 11 Southpoint Business Park
Ensign Way, Southampton
United Kingdom, SO31 4RF
Phone: +44(0) 23 809 870 30
Fax: +44(0) 23 809 870 39
PCE Americas Inc.
1201 Jupiter Park Drive, Suite 8
Jupiter 33458 FL